Assessment of intracochlear trauma caused by the insertion of a new straight research array
- Publication Type:
- Journal Article
- Cochlear Implants International, 2012, 13 (3), pp. 156 - 162
- Issue Date:
Files in This Item:
|Assessment of intracochlear trauma caused by the insertion of a new straight research array.pdf||Published Version||576.72 kB|
Copyright Clearance Process
- Recently Added
- In Progress
- Closed Access
This item is closed access and not available.
Objective: To assess the degree of intracochlear trauma using the Cochlear™ Straight Research Array (SRA). This electrode has recently been released by Cochlear™ on the CI422 implant. Background: Electroacoustic stimulation (EAS) enables recipients to benefit from cochlear implantation while retaining their natural low-frequency hearing. A disadvantage of short EAS electrodes is that short electrodes provide limited low-frequency stimulation. Thus, loss of the residual hearing may require reimplantation with a longer electrode. In order to overcome this problem, the slim diameter SRA with increased length (20-25 mm) has been designed to provide a deeper, yet non-traumatic insertion. Methods: Two insertion studies into temporal bones were undertaken. The first involved dissection of the cochlea to gain a view into the scala vestibuli and insertion of the SRA and control electrodes with a microactuator for a surgeon-independent yet controlled insertion. High-speed photography was used to record data. The second study involved a high-resolution X-ray microcomputed tomography (microCT) study to assess electrode placement and tissue preservation in surgeon-implanted bones. Results: The SRA had a smooth insertion trajectory. The average angular insertion depth was 383° when inserted until resistance was encountered, and 355° if inserted to a predetermined mark for EAS use. In addition, microCT data showed that this caused no significant trauma or distortion of the basilar membrane up to 20 mms depth. Conclusion: Temporal bone studies show that the SRA appears to cause no intracochlear trauma if used as an EAS electrode up to 20 mm depth of insertion. © W.S. Maney & Son Ltd 2012.
Please use this identifier to cite or link to this item: